The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: Tony Williams
Subject: Re: micro ohms measurementsWhat
Date: Sun, 08 Sep 2002 11:18:44 +0100
References: <email@example.com> <firstname.lastname@example.org>
NNTP-Posting-Date: Sun, 8 Sep 2002 10:18:47 +0000 (UTC)
User-Agent: Pluto/1.14i (RISC-OS/3.60)
In article ,
N. Thornton wrote:
> Just a couple of points to add to the thread. Firstly electrode
> resistance is not an issue with 4 wire measurements, and so doesn't
> need any compensating for. That assumes your measuring electrodes are
> sensed by a reasonably high R circuit.
> Secondly a transformer in the measuring side would draw current from
> the measurement elctrodes, thus making the result dependant on
> resistances in that circuit. For that reason it sounds like not the
> way to go. The sense electrodes really need relatively high impedance
> measurement to remain resistance independant.
Yes and no. The reason that many of the classic
bridges are classic is that they are null-seeking.
When this is done the input impedance of the detector
has no effect on the linearity of the bridge, it merely
affects the final resolution. Null-seeking also means
that the linearity of the detector has little consequence.
So the trick here would be to have a transformer, but
have a scheme that always runs the transformer with zero
input. This could either be an actual balancing of the
bridge, or a faked balance with (say) an opposing voltage
in series with the primary of the transformer.
A transformer is desirable, because the step up allows
lower noise and the isolation reduces earth-loop problems.
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup