The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
NNTP-Posting-Date: Sun, 06 Oct 2002 18:27:12 CDT
From: email@example.com (John Fields)
Subject: Re: Strain gauge measurement circuit design
Date: Sun, 06 Oct 2002 22:55:25 GMT
Organization: Austin Instruments, Inc.
X-Newsreader: Forte Agent 1.5/32.451
X-Abuse-and-DMCA-Info: Please be sure to forward a copy of ALL headers
X-Abuse-and-DMCA-Info: Otherwise we will be unable to process your complaint properly
On Sun, 6 Oct 2002 21:31:34 +0000 (UTC), "Simon Price"
>I am trying to construct a circuit that will enable me to measure the output
>from a full bridge strain gauge (Configured as wheatstone bridge).
>The slightly unusual requirement for this design is that I need the system
>to be mobile and be able to re-calibrate itself automatically in a new
>position ( and orientation - within reason).
>I have built a circuit that uses a digital pot to initially 'balance' the
>bridge to give a zero output. The signal from the strain gauge is then fed
>into a high gain amplifier stage and sampled by the ADC on a PIC MCU. The
>PIC also controls the digital pot in order to balance the strain gauge. The
>problem that I have is that the digital POT only has 1024 tap off points
>(The highest I've been able to find) on it and this is causing me issues
>when I try and balance the output as the step size is too large and the
>output from the amplifier stage just swings from one extreme to the other.
>Anybody out there got any thoughts / ideas on possible solutions? Anybody
>tried to do this before?
>Any help gratefully received
Use two pots:
Professional circuit designer
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup