The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: "Spehro Pefhany"
Subject: Re: Strain gauge measurement circuit design
User-Agent: tin/pre-1.4-19990216 ("Styrofoam") (UNIX) (Linux/2.2.14 (i586))
Date: Sun, 06 Oct 2002 23:25:13 GMT
NNTP-Posting-Date: Sun, 06 Oct 2002 19:25:13 EDT
The renowned Simon Price wrote:
> I have built a circuit that uses a digital pot to initially 'balance' the
> bridge to give a zero output. The signal from the strain gauge is then fed
> into a high gain amplifier stage and sampled by the ADC on a PIC MCU. The
> PIC also controls the digital pot in order to balance the strain gauge. The
> problem that I have is that the digital POT only has 1024 tap off points
> (The highest I've been able to find) on it and this is causing me issues
> when I try and balance the output as the step size is too large and the
> output from the amplifier stage just swings from one extreme to the other.
> Anybody out there got any thoughts / ideas on possible solutions? Anybody
> tried to do this before?
Why don't you leave the bridge unbalanced and just subtract the signal
digitally? Alternatively (let's say you are only looking at a tiny part
of the signal so your A/D resolution/range would be inappropriate), you
could use a second digital pot as a "fine" adjustment (you can probably
daisy chain the SPI ports so it doesn't take any more port pins). You
bring it "close" (let's say within 1%) with the fine adjust in a fixed
position and using the coarse adjust, then bring it
in the rest of the way with the fine adjust (which has, say, a 2% range,
giving you steps of 0.002%) You might have to take a close look at the
stability of the various components.
"it's the network..." "The Journey is the reward"
firstname.lastname@example.org Info for manufacturers: http://www.trexon.com
Embedded software/hardware/analog Info for designers: http://www.speff.com
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup