The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
NNTP-Posting-Date: Wed, 09 Oct 2002 22:41:27 CDT
From: email@example.com (John Fields)
Subject: Re: Digital Meter Scaling Question
Date: Thu, 10 Oct 2002 03:46:11 GMT
Organization: Austin Instruments, Inc.
X-Newsreader: Forte Agent 1.5/32.451
X-Abuse-and-DMCA-Info: Please be sure to forward a copy of ALL headers
X-Abuse-and-DMCA-Info: Otherwise we will be unable to process your complaint properly
On 10 Oct 2002 02:35:12 GMT, firstname.lastname@example.org (Ed Edmondson)
>Tom & Others,
>What I need to do is change the meter scaling for a power supply I have built.
>I have two Datel DPM's on the power supply. A 4.5 digit to measure voltage
>(1.25 to 25 volts) and a 3.5 digit to measure current. The actual connection of
>the meters was provided by Datel apps engineers. The volt meter works as it is
>supposed to. It is the current meter which has a problem.
>I am measuring the current being drawn by the power supply by measuring the
>voltage drop across a 0.1 ohm 1% 5 watt kelvin resistor located on the negative
>side of the power supply output (per Datel). At the moment 1 amp reads X1.00
>volts. I want the meter to read 1.XXX amps. If my meter was a 200 mV FS meter
>there would be no problem. However, with a 2 volt FS meter the reading is not
>as accurate (correct)?
>I have some Analog Devices AD8551R op amps as previously suggested in this
>thread. I have connected pin 2 (- In) though a resistor to ground and from pin
>2 to pin 6 (output). The voltage from my current sensing resistor is applied to
>pin 3 (In +). Pin 5 (V plus) to pin 7. Pin 4 (V minus) to ground. The output
>from the op amp goes to the In High pin on the meter and ground goes to In Low
>on the meter.
>I was going to use a 100K resistor from In minus on the op amp to the output
>(feedback) and another resistor 10K from the same In Minus pin to ground. Would
>this be correct?
No. For a non-inverting opamp with the configuration you describe
the gain is described as Vout = ((R1+R2)/R1) * Vin, where R1 is the
resistor from the - input to gnd and R2 is the resistor from the
output to the - input. For a gain of 10, then, the resistance of R2
should be 9 times the resistance of R1.
>What size bypass resistor should I use on the op amp power pin? Is 0.1 or 0.001
Bypass resistor? You probably don't need one, but if if you mean
capacitor, a 0.1µF ceramic Won't hurt.
>What kind of accuracy can I expect using 0.1 % metal film resistors?
1%, since you're using a 1% shunt.
>I realize I could just change the value of the sense resistor or the meter
>itself. However, I already have these components and they are working so it
>would be more trouble than just adding a op amp, 2 resistors, and a capacitor
>between the sense resistor and the meter input.
Professional circuit designer
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup