The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: email@example.com (Bill Sloman)
Subject: Re: Cheap low frequency impedance bridge?
Date: 11 Oct 2002 14:55:36 -0700
NNTP-Posting-Date: 11 Oct 2002 21:55:37 GMT
John Popelish wrote in message news:<3DA62DF9.E6567ECA@rica.net>...
> Bill Sloman wrote:
> > I've got a component at work that looks like a resistance in series with a
> > capacitor, and I've got to monitor the capacitance fairly precisely to look
> > for degradation in use - the device is set up so that the resistive
> > impedance (which is what we are interested in) almost dominates the
> > impedance at the measurement frequency, but the capacitance can get eroded
> > in service, and we are going to have put numbers on this.
> > The measurement frequency range is 2kHz to 200kHz, and we'll probably
> > monitor down to rather lower frequencies if we can, to boost the reactive
> > impedance to more or less match the resistive component.
> > So I need an impedance bridge. I don't need - and couldn't pay for - an RF
> > mpedance bridge going into the MHz range.
> > The resistive impedance range runs from to 10R to 10k, but the region up to
> > 100R is the most interesting. The capacitance runs from about 1uF to about
> > 100uF, and the higher values are frequency dependent (just to make life
> > interesting and to exclude LCR measuring gear that doesn't give you control
> > over the excitation frequency).
> > The excitation voltage should be about 5mV rms - we've used more (up to
> > about 100mV) without seeing any gross problems, but more precise
> > measurements may need the lower drive level.
> > We've done some preliminary work on an HP 4192A bridge in a nearby
> > university lab, at vast expense. The HP4192A was just what I wanted, but
> > Agilent don't make it any more, and my boss couldn't afford to buy it if
> > they did.
> > Anybody know of anything cheap that offers frequencies up to 200kHz? I used
> > a signal generator and an oscilloscope for the intial development, a couple
> > of years ago, but for this work we really need something rather more
> > precise.
> > ----
> > Bill Sloman, Nijmegen
> Used General Radio 650A and 1650A impedance bridges go pretty cheaply
> on EBAY all the time. I don't know how good they are at 200 kHz, but
> they work well over the audio range, and at least have calibrated
> dials. Don't know how much sensitivity they give at 5 mv excitation,
> but I think you can amplify the signal that goes to the null meter for
> more sensitivity.
There is a 650A on e-bay for $19.95, but I couldn't get any details on
it. My boss likes the price. A classical bridge could well be an
interesting option, but I might have to download Jim William's low
distortion oscillator from the Linear Technology web site if I wanted
a really good null.
Bill Sloman, Nijmegen
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup