The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: Winfield Hill
Subject: Re: Cheap low frequency impedance bridge?
Date: 17 Oct 2002 15:03:17 -0700
Organization: Rowland Institute
References: <firstname.lastname@example.org> <email@example.com> <firstname.lastname@example.org> <email@example.com> <firstname.lastname@example.org> <email@example.com>
X-Newsreader: Direct Read News 2.98
> I had to ring Thurlby-Thandar to get the excitation voltage - they
> didn't respond to my e-mail - and at 30mVrms it is more than I would
> have liked, but less than the 50mV rms "low" level from the HP 4276
> that Win recommends.
You need a grown up LCR meter, like my HP4275A. This has adjustable
excitation in three ranges from 0.01V to 1V. On the lowest range I
turned the control pot down to 0.1, the excitation voltmeter read 1mV
but the capacitance reading didn't change at all. Nice. I then turned
the excitation control pot to zero and the meter read 0.000V, yet the
4275A still gave the correct cap reading. Now that's impressive :-)
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup