The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: email@example.com (Kooner)
Subject: Re: question on measuring temperarures at high voltages
Date: 23 Oct 2002 10:10:00 -0700
NNTP-Posting-Date: 23 Oct 2002 17:10:00 GMT
"Ban" wrote in message news:...
> "Kooner" schrieb im Newsbeitrag
> > I tried to measure the temperature of a LT1084 voltage regulator that
> > is floating at 10kV. The 10kV is the output of a switching power
> > supply. The regulator is in a TO-247 package. I placed a thermocouple
> > on the body, which is supposed to be isolated from the leads and case
> > (which are at high voltage). The thermocouple was plugged into a hand
> > held temperature meter powered by a 9V battery.
> > My question is why does the meter give an invalid reading? The
> > temperature just bounces around. I'm guessing it's one of two things:
> > 1. The thermocouple acts as an antenna picking up EM fields that
> > induce a voltage.
> Yes, this will probably be the reason. When switching off the circuit to be
> measured do you get a stable reading?
Yes I get a stable reading when I shut the power off.
> > 2. The body is not totally isolated from the high voltage leads.
> > Therefore the 10kV forces a current along the body, to the
> > thermocouple, to the hand held meter to ground. But the hand held
> > meter is not referenced to ground??
> When the meter is placed on an isolating surface(rubber mat, dry wood,
> plastic) it will still work, as long as there is no connection to mains. If
> there is it will never work again.
> I wouldn't hold that meter in the hand! consider that a 10kV potential can
> bridge a substantial distance of air and zap...
> do not fiddle around with voltages that high without a decent training.
> There are temperature sensitive self adhesive plastic dots available, which
> change the color when the set temperature is exceeded.
> There are also infrared sensors for contactless temperature measurements.
> Since your application doesn't require the accuracy of a thermocouple, be
> smart and use one of those.
Well, I do have a little training working with high voltage. I'm a
junior EE studying/measuring someone else’s design of a 10kV
power supply used to power a traveling wave tube (TWT). Our end
product is a microwave amplifier.
The LT1084 is used to power the filament or "heater" in a TWT. I can
power up the LT1084/filament circuit without turning on the high
voltage, and safely measure the power dissipation at 4.5W at worst
case and a case temperature at 115C. The LT1084 is also mounted to a
small crappy heat sink and there's no room for a better one.
The problem is I need to measure the temperature of the LT1084 in the
end product - at high voltage. Thermal dots have a 15C window which
in not accurate enough. Infrared sensors can be better if they have an
emissivity setting on them. But I couldn't use it anyway because the
LT1084 is part of a high voltage module that gets incased in a red
block of potting material to prevent arcing and corona. The block of
potting material has other power dissipating components in it also has
a thermal conductivity of 1 W/(mK) which is a lot better that air. I
can estimate the LT1084 temperature, but with these other variables it
really would be guess. My guess is it gets to hot, but I can't prove
it without a valid measurement.
Wasn't there a famous scientist who said, "We know what me measure."
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup