The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: Chuck Simmons
Organization: You jest.
X-Mailer: Mozilla 4.61 [en] (X11; U; Linux 2.0.33 i586)
Subject: Re: vibration measurement circuit (vibration gurus please help)
References: <6EGu9.email@example.com> <3DBB5EB6.D2A4AECA@webaccess.net> <9kTu9.firstname.lastname@example.org>
Date: Sun, 27 Oct 2002 16:14:00 GMT
NNTP-Posting-Date: Sun, 27 Oct 2002 08:14:00 PST
mook johnson wrote:
> "Chuck Simmons" wrote in message
> > mook johnson wrote:
> > >
> > > I'd like to get a graph of the vibration level (Gs) over a frequency
> > > of 10 to 5kHz
> > > The vibration to be measured would be continuous or slowly changing over
> > > several minutes.
> > > Seven bands over that interval would be sufficient resolution of the
> > > frequency.
> > >
> > > Is it valid to do this using an Analog Devices accelerometer to feed 7
> > > bandpass filters.
> > > then measure the peak amplitude of each of the 7 BP outputs?
> > >
> > > I was thinking of using peak detectors to hold the peak experienced in
> > > band over a 300mS period then use an A2D to sample the peak in each
> > >
> > > I only have an 8051 or MicroChip PIC running at 7.3MHz to do the
> > > A2D manipulation and serial communication to the host at 9600 bps. 3
> > > different axes will be sampled over the 1 second period and the host
> > > request an update at 1 /second.
> > >
> > > I think FFT is out due to limited processing power and limited local RAM
> > > storing the raw waveform.
> > >
> > > Is this a valid approach? Will this be useful data for verifying
> > > qualification vibration levels for a device to be used in the same
> > > environment?
> > >
> > > The desired display on the computer screen would be 3 graphs with
> G(peak) on
> > > the Y axis and frequency on the X axis.
> > The method you mention is valid. I used it back in the early 1970s on a
> > project to detect formation of microcracks in samples during working of
> > the samples to cause fatigue. My frequency range was in the megahertz
> > but the method makes sense. However, with a full second to do
> > processing, I would consider some sort of DFT method, say an FFT. I
> > don't know how much RAM you have but if you have enough, the processing
> > might fit in the time. The embarrassing thing is you want 10Hz because
> > that means that you need to sample at 10kHz and collect 1024 points for
> > a simple FFT. If the processor can have external RAM and careful coding
> > will meet time constraints, use the processor. Otherwise go with the
> > filters. If you do the digital approach, you will have to have an
> > antialiasing filter.
> > For either approach, you will need to use a shaker table for
> > calibration.
> > Chuck
> We only have about 128 bytes of SRAM available with the 8051 and 350 bytes
> with the PIC.
> Not enough room for storage of the waveform.
Analog filters then unless you can find a microP with enough memory or
can do external RAM.
... The times have been,
That, when the brains were out,
the man would die. ... Macbeth
Chuck Simmons email@example.com
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup