The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: Paul Burridge
Subject: Re: Amplifying stage with *negative* Voltage gain??
References: <email@example.com> <3DEF1DC0.5EDD@Spam.Bots> <firstname.lastname@example.org> <3DEF5BEA.2614@Spam.Bots> <3DF0A422.46EE@Spam.Bots> <email@example.com> <3DF263CE.3868@Spam.Bots> <firstname.lastname@example.org> <email@example.com> <3DF61BF3.5430@Spam.Bots>
X-Newsreader: Forte Agent 1.8/32.548
Date: Tue, 10 Dec 2002 14:06:29 +0000
NNTP-Posting-Date: Tue, 10 Dec 2002 15:34:51 GMT
Organization: Virgin Net Usenet Service
On Tue, 10 Dec 2002 08:53:07 -0800, Mike Monett opined
>The standard scope input is 1 meg in parallel with 30 pf or so. You can
>measure it with a dvm to verify, but it is not likely to be off by much.
>All your readings would be out of calibration with a 10X probe.
Confusing. Out of calibration makes the readings inaccurate; using the
X1 probe makes the readings inaccurate. Can't win! :-(
>You can also measure the resistance of the probe. On the 1X setting, the
>resistance should be low (short-circuit). On the 10X setting, you should
>see 9 megs.
I do indeed, but curiously, the final reading ended up swinging
periodically up and down at around 9M every few seconds. Never seen
such behaviour from a passive device before. Unless it's something to
do with the action of the meter. I'm beginning to question the
God-like faith I had in measuring equipment. :-(
>As usual, Tony is pointing us in the right direction. The symptoms
>indicate some large capacitive load on the output. The trace capacitance
>will be in the pf range, not enough to affect things. Can you remove the
>1 nF output cap and measure directly at the pulldown resistor?
See my other reply to Tony.
>Also, using a lower frequency might help in debugging. If the waveform is
>clipped at 1 KHz, there is definitely some new physical process going on
>considering the voltage measurements you posted previously.
I did try to pop out and borrow a sweep gen. earlier but my car locks
were frozen solid. That little test will have to wait...
>There might be a Nobel prize... - nah, probably not!
Only for sheer dogged determination! (on the part of all of us)
"I was going to help him up, but then I thumped him."
Michael Head's plea in mitigation to Weybridge Magistrates, Sept. 1975
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup