The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: "Da Man"
References: <firstname.lastname@example.org> <email@example.com> <3E100CF4.1AB@sneakemail.com>
Subject: Re: Total lack of response re SMT query - What gives ?
X-Newsreader: Microsoft Outlook Express 5.50.4807.1700
X-MIMEOLE: Produced By Microsoft MimeOLE V5.50.4910.0300
Date: Tue, 31 Dec 2002 23:51:39 GMT
NNTP-Posting-Date: Tue, 31 Dec 2002 19:51:39 AST
"Mike Monett" wrote in message
> Nico Coesel wrote:
> > Serious now. Even those small components can be soldered with a small
> > soldering iron. Maybe you need a special pair of glasses for fine work
> > (consult an optometrist) and a good lamp. A stereo microscope is handy
> > for inspection work, but the size of the image is probably far too
> > small to assemble a PCB.
> > Reply to nico@nctdevpuntnl (punt=.)
> A low power 7X to 30X stereo zoom with bright illumination is invaluable
> for checking the quality of solder joints. The problem is the stage is
> too small.
> I bolted a 1/8" piece of aluminum 13" X 20" to the stage, with threaded
> standoffs to support the edges. The top is covered with a thin sheet of
> material used for kitchen countertops, and horizontal and vertical
> crosshairs are scratched into the material intersecting at the focus. I
> can place a rather large pcb on the stage and use the crosshairs to align
> the part I'm working on with the microscope focus area.
> Illumination is provided with two miniature 40 watt stage bulbs
> controlled with an ordinary light dimmer. A small aluminum shield is
> provided to keep my wrists from touching the bulbs while adjusting the
> microscope. I'd like to use small flourescent bulbs to reduce the heat,
> but it's difficult to adjust the light level. Another alternative might
> be to find illuminators using fiber optic light pipes, but watch out for
> the lamp replacement cost.
Try a ring made of white LED's?
> Over the years, the glue holding the lens dried out and the lens fell
> out. I was afraid of taking the microscope apart due to possible damage,
> but found the process quite simple. A small dab of automotive high
> temperature silicon rubber with low outgassing for oxygen sensors solved
> the problem.
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup