The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
From: "Al Hephy"
Subject: Re: Humidity sensor - peculiar behavior
Date: Mon, 6 Jan 2003 09:06:59 -0500
Organization: Posted via Supernews, http://www.supernews.com
Reply-To: "Al Hephy"
References: <email@example.com> <firstname.lastname@example.org>
X-Newsreader: Microsoft Outlook Express 5.00.2615.200
Steve Turner wrote in message =
> Many thanks to those who have contributed ideas for solutions to this
> I believe I have eliminated the possibility of a faulty sensor. I
> have removed the sensor from the circuit board and suspended it naked
> in the 75% constant humidity chamber. The reading is spot on with
> only negligible drift.
> Oddly, though, I cannot reproduce the drift problem by plugging the
> end of the sensor cable into same circuit as exists on the printed
> circuit board. This configuration also gives a stable reading! The
> only difference here is that the circuit is on a solderless breadboard
> outside of the chamber instead of on a printed circuit board inside
> the chamber.
> I was able to recreate the original problem by soldering the sensor
> back into the circuit board. Inserting a 100 ohm series resistor
> between the sensor output and the op amp input, and a 0.1 uF cap
> between this input and ground, may have slowed down the output voltage
> decay, but the output still ends up far out of tolerance from where
> it's supposed to be.
> I'm stumped. I guess I'll power the sensor only when actively taking
> a reading.
> Again, thanks to all who replied. I appreciate the help.
> Steve Turner
> Real address contains worldnet instead of spamnet
The experiments you describe above would imply that your PCB is =
sensitive to humidity...
I'd look very carefully at the input to the op amp.
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup