The Cyber-Spy.Com Usenet Archive Feeds Directly
From The Open And Publicly Available Newsgroup
This Group And Thousands Of Others Are Available
On Most IS NNTP News Servers On Port 119.
Cyber-Spy.Com Is NOT Responsible For Any Topic,
Opinions Or Content Posted To This Or Any Other
Newsgroup. This Web Archive Of The Newsgroup And
Posts Are For Informational Purposes Only.
Subject: Re: Water sensor
References: <email@example.com> <firstname.lastname@example.org>
X-Newsreader: Forte Agent 1.91/32.564
NNTP-Posting-Date: Tue, 14 Jan 2003 16:15:06 GMT
Organization: Insight Broadband
Date: Tue, 14 Jan 2003 16:15:06 GMT
On Tue, 14 Jan 2003 09:20:57 +0000 (GMT), Tony Williams
wrote in message
>In article ,
> Bruce Murphy wrote:
>> Tony Williams writes:
>> > When in water the thermistor will struggle to get
>> > above 100C as it tries to evaporate the water off
>> > (540 cals/gm?). At that point its resistance will
>> > be stuck down at the 100C value.
>> While this is a cute idea, since the sensors have to be small, this
>> strikes me as replacing one hazard with another.
> It's a cheap (and cruder) version of what is
> used for oil-level sensing in gearboxes, etc.
> In that app the thermistor resistance is measured
> at low power, then it is given a high power for
> a fixed time, and the resistance measured again.
> (T1 - T2) determines whether the thermistor is
> immersed in oil or in air. Widely used in the
> automotive and avionics industries... there are
> even ic's to do the job.
You also need to measure temperature because thermistors are 'notorious' for
changing value with temperature ;-) One way is to use two thermistors, one of
which is always submerged (or not) and in same temperature/heat conductance
In 1985 or so I presented a paper on a thermistor-based method to assess oil,
water and air saturation in porous media such as sand based on both static and
heat pulse measurements. The latter provides a waveform which contains more
information for interpretation than two static measurements.
Do you know of a replacement for the now-discontinued LM1820 Fluid Detector IC?
Go Back To The Cyber-Spy.Com
Usenet Web Archive Index Of
The sci.electronics.design Newsgroup